MJ-560B2 anesthesia machine
MJ-560B2 anesthesia machine’s design is union the modern advanced technology and with its strong practicality, which is a machine of excellent performance and fully functional. The remarkable performance, the preferential benefit price, constitutes the best performance-to-price ratio, causes the user to obtain the over ＆#118alue to enjoy.
■ This machine uses in respiratory control , respiratory intensive care , 5.1-inch high clear display, and real-time reflection respiratory system condition;
■The high-precision evaporator has the stable output concentration with pressure, temperature, and flow automatic compensation; Choose one among Enflurane, isoflurane and sevoflurane for a matching.
■Reliable gas Power transmission system;
■ Reasonable economic and multiple technology configuration, the user s may choose flexibly;
■ modular product design According to customer’s needs upgrade. Main technical parameters Part of Anesthesia Machine
■Console High-strength engineering plastic rack, light, beautiful and corrosion resistance
■scope Adults, Children
■Gas source oxygen、nitrous oxide:0.3~0.5MPa
■flow meter oxygen: 0.1~10L/min nitrous oxide: o.1~10L/min
■oxygen、nitrous oxide linkage and nitrous oxide Truncation device When use nitrous oxide, oxygen concentration>25%; when the oxygen pressure<0.2Kpa, the flow of nitrous oxide would be cut off.
■Rapid oxygen flow 35~75L/min
■Vaporizer It has the function of pressure, temperature, and flow automatic compensation. The regulation range of evaporator concentration is 0~5vol%. Choose one among Enflurane, isoflurane and sevoflurane for a matching according to the demands of customer.
■Low oxygen pressure alarm It will be alarmed when the oxygen pressure < 0.2MPa
■respiratory circuit Work mode all close. semi-close. semi-open Pressure release 1~4KPa
■Bellows breathing Adults and children bellows Tidal volume range:0~1500ml Part of Ventilator
■Display mode 5.1-inch high clear display,
■Ventilation mode IPPV SIPPV Artificial monitoring MANU(monitor the tidal volume. ventilation. respiratory rate. ratio. peak pressure when use hand.)P PEEP
■Ventilation parameters Tidal volume 50~1500ml Frequency 6~60/min breath /expiratory ratio 2:1~1:4 Trigger sensitivity -1.0~-0.2Kpa PEEP 0.2~1.0Kpa Pressure Control 1~6.0Kpa
■Ventilation monitoring parameters tidal volume. ventilation. IPPV ratio. total respiratory rate. breath /expiratory ratio. peak pressure . pressure control. airway pressure wave. breath pressure trigger. PEEP. minimum pressure .
■Monitoring oxygen concentration 21%~100%(choose and match)
■Security Alert System Oxygen Concentration alarm upper limit setting range 50%-100% (choose and match) low limit setting range 15%-50% Airway pressure alarm upper limit setting range 1.5~6.0Kpa low limit setting range 0.2~3.0Kpa Alarm minute ventilation upper limit setting range 2~30L/min low limit setting range 1~12L/min Asphyxia alarm The machine will give sound and light alarm if there is no tidal volume input in 15s Power failure alarm
■ Power AC 220V±10% 50Hz±1Hz UPS. Battery. (choose and match)
Send Inquiry to this Member
Related Products of this Company
This member assumes full responsibility for the content of this listing. DIYTrade accepts no responsibility whatsoever in respect of such content.
To report fraudulent or illegal content, please click here
China Suppliers Quick Searching: