|Origin:||Made In China|
|Category:||Electronics & Electricity / Electronic Instrument / Electronic Measurment Apparatus|
|Label:||laser rangefinder , rangefinder , laser range finder|
|Min. Order:||3 pc|
China TianYing TY-LR93 miniaturized 40km laser rangefinder module 20 Hz for 5m² plane 20km, designed for integration into modern optronic reconnaissance tracking systems, provides accurate range performance to fast moving targets in extremely compact unit that weighs 5kg.
Depending on weather conditions and target characteristics, ranges between 200 m and 50 km can be measured. The measuring accuracy is 1m. Up to 500 measurements can be carried out in 25 seconds with a measurement frequency of 20 Hz. The model also has a 1570nm vision, no increase in size or weight.
The LRF-40km leveraged the lastest technology to construct this remarkably compact, exceptionally accurate rangefinder, hence customer enable very easy integration into ships, land vehicles or aeroplanes platforms, inclduing camera system, fire-control system etc.
40km 20Hz continuous Laser Rangefinder Features:
1) Up to 50 km ranging capability, typical 5m² plane >18km and up to 20km of 20Hz(tank >15km).
2) Pulse repetition frequency up to 20 Hz continuous.
3) Range accuracy < +/-1 m for target ranges.
4) Diode-Pumped ND:YAG, option OPO.
5) ≤ 300 μrad beam divergence.
6). TE-cooled avalanche photodiode detector.
7). Miniaturized design, weight < 5 kg with house.
40km 20Hz continuous Laser Rangefinder Module Recommended Use:
1) Optronic surveillance and reconnaissance.
2) Target tracking systems and anti-aircraft systems.
3) Fire control systems for land, air and sea targets.
4) For determination of flight altitudes on airports.
5) To measure the distance of ships from the coast.
6) In general for applications of position determination of moving objects.
40km 20Hz Laser Rangefinder Module mainly specification:
|Laser Type||Diode-pumped +Nd:YAG|
|Wavelength||1064nm, option 1570nm|
|Measuring rate(programmable)||Typical 20Hz work 25 seconds and rest ≤ 25 seconds cycle .|
|Beam divergence||≤ 0.3 mrad.|
|Boresight stability||≤ 0.05mrad.|
|Cooling||Fan air flow.|
|Detector||TE-cooled Avalanche Photo Diode Detector (TE-cooled APD).|
|Max Ranging capability||40km of building/large craft in 45km visibility;|
|> 18km of 30% albedo 5m² aircraft in 23.5 km visibility(tank >15km).|
|Power||20 - 30V DC|
|Life||≥ One Million Range Measurements|
|MTBF||> 10000 hours|
|Size||≤ 6000 cubic centimeters|
|Weight||≤ 5000g with house|
|* - Note||*Product design and Technical information is subject to change without notice
** Depending on the region of sale.
1, Why many 20km/30km/40km laser rangefinder can not measure tank 10km in standard 23.5km visibility?
Because their 20km/30km/40km is theory data or high reflection large target in ideal visibility condition, even reflection need beyond 80% and visibility need beyond 50km.
2, Why many NATO target 10% reflection 12km laser rangefinder can not measure tank 10km in standard 23.5km visibility?
Because their 10% reflection is not real performance data, 2.3m x 2.3m NATO target is not real tank, or their NATO target effective size is different; normal 2.3m x 2.3m target real albedo is 50% or more.
About test land standard, normal big target typical 20km LRF with 25km visibility, then, normal this LRF can measure 2.3m x 2.3 target about 15km or more, then, 30% albedo 2.3m x 2.3m target is about 10km ~ 12km.
Also, starting state and stable state measuring range is different too, such as 15mJ-25mJ LRF, starting state is 25mJ, typical stable state is only 15mJ, only many companies are only said starting state, for example, said single pulse laser energy is at least 25mJ.
3, Why many laser rangefinder do not show measurement rate details?
Because max rate and continuous rate technology demand is different, such as max 12.5Hz 10 sec to 60 sec, maybe then 5Hz is max 1 min to 3 min with cycle.
Also, when rate have more different, such as 1Hz-5Hz, 1Hz-10Hz, 1Hz-20Hz, 5Hz-25Hz etc, then their working distance have more different too.
About 1.5µm eye safe laser rangefinder, if 1Hz is 10km, then, normal 5Hz is only balance about 80%~ 85%, normal 10Hz is only balance about 70%~80%, when this LRF laser energy is very low, then attenuation speed will accelerate.
4, Why normal mid-long range high rate laser rangefinder do not use for Erbium-glass laser?
Because normal Erbium-glass laser energy is too low, high rate continuous measuring time is too short, especially at extreme operating temperatures.
In fact, if Erbium-glass laser want to meet demand, energy need at least 2mJ, normal need 5mJ to 8mJ.
Then, Erbium-glass is not competitive with OPO and Raman shifted.
5, Why many laser rangefinder price is very cheap?
Because performance is different, for example: one LRF is used for bad cable, messy interface, messy wiring etc, then the LRF cost can reduce, even reduce much, but no warranty on performance.
If used for same performance, then our price is very cheap. that is why many laser rangefinder do not show their real main performance data.
|Embed HTML Code:||
HD800 2MP 12.5mm~800mm Coaxial Optical Zoom Surveillance Security Camera
HD800PRO 2MP 12.5mm~800mm Coaxial Optical Zoom Surveillance Security Camera